|Place of Origin:||China|
|Minimum Order Quantity:||1 unit|
|Packaging Details:||Carton box|
|Delivery Time:||7 working days|
|Payment Terms:||Western Union, MoneyGram, T/T|
|Supply Ability:||10 unit / month|
|SKYLINE:||Creases Recovery Tester||Model:||SL-F08|
|Size Of Test Specimen:||40 X 15 Mm||Teasing Load:||1.0 Kg|
|Angle Measurement:||10 To 180degree X 10||Standard:||ASTM D 1295-1967|
textile lab equipment,
textile testing instruments
Creases Recovery Tester
The ability of textiles to recover from creases is determined by measurement of its crease recovery angle. In this test the test specimen is first creased under a specified load for specified period of time and it then transferred to the angle measuring device where one of its ends is held in a spring loaded clamp while the other end is allowed to fall freely under its own weight. This clamped end is slowly rotated till the free end becomes vertical. The angle of deflection of the clamped end from the horizontal is read on a scale fixed to the ability of textiles to recover from creases is determined by measurement of its crease the moving clamp. This angle gives the measure of the crease recovery.
Crease Recovery Tester for fabrics consists of an arrangement to crease a rectangular test specimen under a specified load for the specified duration of time and an arrangement to measure the crease recovery angle. Creasing is carried out by pressing the folded tests specimen under a load of 1 kg between two flat surfaces. The load is applied by mean of the dead weight of the upper flat surface.
The arrangement for measurement of crease recovery angle has a anodised aluminium scale calibrated in degrees mounted on a vertical disc. The disc can turn about the axis of the scale and has a spring loaded clamp located at 00 to hold one end of the test specimen. The free end of the test specimen can be aligned against a vertical edge fitted in front of the disc.
The test specimen is mounted on a brass strip for convenience in use and is creased and held in the clamp while still fixed onto it.
|Size of test specimen||40 x 15 mm|
|(Teasing load )||1.0 kg|
|Angle measurement||10 to 180degree x 10|
|IS 4681 - 1968||Method for Determination of Wrinkle Recovery of Fabrics (By measurement of Crease Recovery Angle)|
|BS 3086 - 1972||Determination of Wrinkle Recovery of fabrics. Crease Recovery Angle Method.|
|ASTM D 1295-1967||Standard Method for Wrinkle Recovery of Woolen Textile Fabrics using the vertical strip apparatus.|
Contact Person: Ms. Ivy Xie