|Place of Origin:||China|
|Minimum Order Quantity:||1 unit|
|Delivery Time:||5 working days|
|Payment Terms:||T/T, Western Union, MoneyGram|
|Supply Ability:||1 unit|
|SKYLINE:||Sharp Point Tester||Standards:||ISO 8124-1 EN-71,ASTM F963|
|Weight:||1.5kg||Dimensions:||150*100*60 MM (L*W*H)|
|Accessories:||Weight, Batteries, Bulb 2 Pieces||Warranty:||18 Months|
stability test chamber,
swings testing equipment
ISO8124-1 Toys Testing Equipment Sharp Point Tester
To determine whether accessible sharp points on toys are likely to cause injury. If the accessible sharp point penetrates a specified depth into the small rectangular opening of the tester, the LED is illuminated to indicate that the point is unacceptably sharp. Toys intended for use by children under 8 years old are subject to this requirement before or after use & abuse testing.
Accessories: weight, batteries, bulb 2 pieces
1. Place the sharp point tester on the accessible point and check whether the tested point can be inserted into the sharp tester to the specified depth. The pointed point of the test is inserted into the measuring slot, and an external force of 1 pound is applied to make the indicator light on. This pointed point is judged as a sharp pointed point.
2. Wooden thorns in wooden toys are dangerous sharp points, so they must not be present on toys.
3. Before each inspection, the sensor head must be adjusted according to the regulations to ensure accurate and sensitive sensing.
4. When adjusting the sharp point tester, first loosen the locking ring, and then rotate the locking ring to move it forward enough to the indicator device to expose the calibration reference scale on the circle. Turn the measurement cover clockwise until the indicator light flashes. Rotate the measuring cap counterclockwise until the appropriate micrometer mark is consistent with the calibration scale, and then turn the locking ring until the locking ring is against the measuring cap to fix the test cap in place.
16 CFR 1500.48, ASTM F963 4.8, EN-71 1998 8.14, ISO 8124
POWER 1.5AA batteries x 1 (for 2000 times test)
|DEMENSIONS 150*100*60 MM (L*W*H)|
What is sharp point tester?
Sharp Point Tester is used to test if test object point is sharp or not. Sharp Point Tester in textile, footwear is commonly used to test ornament point sharpness. It is one of necessary safety testers widely used in toy, textile and other industries.
Contact Person: Kate Guo