1、The automatic micro-moisture tester should be placed in a dry place when not in use.
2、Don't put the instrument to work in the environment with corrosive gas.
3、Do not place the instrument in an environment with high humidity and large power fluctuations to avoid abnormal phenomena.
4、In case of poor sealing effect of the silica gel pad inside the inlet port, replace the silica gel pad according to the instruction.
5, a long time application plug, socket will adhere to the dirt, so that its poor contact, then the application of ethanol to wipe the metal parts respectively to make good contact.
6, the day after the test, be sure to empty the system of Karl Fischer reagent, and then cleaned with methanol, never use water to clean the system, because it is not easy to volatilize, will cause the next test Karl Fischer reagent calibration is not true.
7, automatic micro-moisture tester failure should immediately cut off the power, ask professionals to repair and troubleshoot before continuing to use, to prevent accidents
Contact Person: Ms. Ivy Xie