Kinetic Energy Tester Toys Testing Equipment
This device comes with a printer.
The instrument is mainly used to test the kinetic energy of the projectile toy, so that the ejection component of the toy passes the test tube of the instrument or the two sensors of the external test component, and the time taken by the toy ejection component to pass a certain distance is measured, and the movement is calculated. energy. Inspect the safety performance of the catapult toy for infants or children.
The sample is injected from the front of the test channel, and the instrument automatically captures the velocity of the object and automatically records the time difference between the front and back positions of the test. Based on the measured time value and the weight of the sample, the kinetic energy value of the sample and its hazard can be calculated.
| Number of samples|| 1/time|
| Screen control|| Touch Screen|
| Maximum timing|| 15.000000S|
| Timing range|| 0.000001--15.000000S|
| Speed display range|| 0.007M/S—10000.000M/S|
| Front and rear sensing spacing|| 152.4mm,error0.1mm|
| Minimum diameter of sample|| 3mm|
| Test save|| save data|
| Volume(WxDxH)|| 36x35x15cm|
| Weight (approx.)|| ≈6kg|
| Power supply|| 1∮ AC 220V 3A|
Special Selling point
1. Touch Screen
2. With print
3. The machine will automatically calculate the kinetic energy
4.Test save, can save data
5. Calibration function
6. Price concessions
7. Independent packaging
8. In stock, can be shipped immediately
The packing details is as below :
With the detection capability of bullets above 0.5mm diameter, no sensitivity adjustment is required, and it is fully automated. Enter the weight of the bullet,
Automatic calculation of bullet velocity, kinetic energy, number of times, average
Please review video of the Kinetic Energy Tester operation as below link:
Calibration procedure,the simulator output signal port is well connected with the instrument signal interface.
Simulator time input Dial the dial to set the time. Input power connection +5V/1A.
Need to press in the touch screen
Start the test button. After everything is installed, press the simulator output switch.
The result will be Display the touch screen to see if the input time is the same as the displayed result. (+/-0.0001 second error)
The difference between the two kinetic energy
Touch screen version: Touch screen, upgraded kinetic energy, easy to operate, will record time, speed, Machine automatically calculates kinetic energy, and print and Calibration function.
Regular version:Ordinary kinetic energy, button version, operation is not so convenient, only the time and speed are recorded on the screen, the kinetic energy needs to be calculated by yourself, there is no printing function.